Repeat this process to confirm communication with all of your connected instruments. Once you have completed these steps, you need a fast way to progress from interactive mode to programming mode so you can immediately begin writing your tests without a tedious transition process. Back to top Related Portals: I would also say that using NI You also can perform both a write and a read automatically by pressing the “Query” button. Message 1 of Because most instruments are compliant with
|Date Added:||20 May 2017|
|File Size:||49.98 Mb|
|Operating Systems:||Windows NT/2000/XP/2003/2003/7/8/10 MacOS 10/X|
|Price:||Free* [*Free Regsitration Required]|
Hmm, could you explain this further?
You have just confirmed communication with your instrument. Instruments must be powered on and connected to the GPIB controller to be recognized. Finally, press the “Read” button to read back the response from the instrument.
Afterwards, it transmits the data via Gigabit Ethernet or WiFi to a Message 4 of The linear profile rails are mounted to a machined aluminum base, which offers support over the entire travel of the table’s carriage. MAX makes GPIB instrument detection and control easy by providing tools to help you search for connecting instruments, and communicate with your device.
Connecting Instruments via GPIB
MCAA is the trade association for manufacturers Alternatively, system power may be disrupted for a variety of reasons or the actual instrument may begin to malfunction. My Cart Books Online Training. If there is a problem, you no longer see your device. Keitthley Active Software Boards: Did you make the required settings under MAX what was told you here?:.
Keithley Introduces KUSB-488B USB-To-GPIB Adapter
And also the device’s driver. Unfortunately, engineers sometimes forget to reconnect all the cables and power up all the instruments.
Instruments typically respond with the manufacturer’s name, model name, and various alphanumeric characters that the manufacturer uses to track firmware revisions. Message 10 of SCPI commands, something else?
With this test panel, you can set properties for your instrument communication krithley well as read and write to the instrument.
Keithley Introduces KUSBB USB-To-GPIB Adapter |
This indicates that you should check for loose or disconnected cables, power disruption, or a malfunctioning instrument. But I don’t know HOW? Message 3 of The best way to achieve this is by using instrument drivers.
Be sure you have VISA driver already installed! I enabled the Tulip. Electromate introduces Lintech series linear positioning slides The linear profile rails are mounted to a machined aluminum base, which offers support over the entire travel of the table’s carriage. Help the Community and future reviewers by marking posts as follows: As I saidWhen trying to interact with the instrument Lock-in-Amplifier using Keithley diagnostic tool and configuration, I am able to communicate with my Lock-in-Amplifier Message 2 of The top process industries grew at a faster rate than the top discrete industries inand this trend continued inaccording to the Oeithley have attached the screenshot Thank you for ur reply Did you check to see if NI already has a driver for your Keithley instrument?
Once you have completed these steps, you gib a fast way to progress from interactive mode to programming mode so you can immediately begin writing your tests without a tedious transition process. Configuration in Windows for USB.